Titan Penile Implant Instructions: Inflate/Deflate
Read the penile implant instructions below to learn how to inflate and deflate your Titan® inflatable implant.
The Titan® inflatable implant
To Simulate an Erection
To transfer the fluid to the cylinders, locate the pump in the scrotum. Squeeze the pump firmly a few times between the thumb and fingers until an erection is achieved.
To Return to a Flaccid State
To return the fluid balance to the reservoir, follow the tip of the pump as it narrows until the protruding bars are felt. Position thumb and finger on the release bars above the pump. Place thumb on one bar and a finger on the opposite bar. Squeeze release bars with firm and steady pressure. Continuous pressure is necessary. To achieve a flaccid state more quickly, squeeze the penis gently with the other hand while squeezing the release bars. If any problems are encountered during the use of this device, or any pain or discomfort is experienced, please contact your doctor immediately.